Trimble Thunderbolt monitor, almanac says oké green. UTC is correct but the date hangs at Sept 29 Week number rollover problem. GPSDO functions normal. Low signal less then 30mVtt at the 10MHz port. Place a wire connection to the VCXO out and BNC with 0.1u Now having 500mV tt 10MHz out signal after mod.