Trimble Thunderbolt monitor, almanac says oké green. UTC is correct but the date hangs at Sept 29 Week number rollover problem.
GPSDO functions normal.
|Low signal less then 30mVtt at the 10MHz port. Place a wire connection to the VCXO out and BNC with 0.1u|
Now having 500mV tt
|10MHz out signal after mod.|